RickSFO Senior Member Joined: 30 Dec 2012 Posts: 3 Location: San Francisco Expertise: Professional
Posted Sat Apr 20, 2013, 10:19am Subject: Re: coffee refractometer
On the VST LAB refractometer, does anyone know what the "temperature stability function" is? It appears to be different from the temperature correction since it's listed as a separate feature. Perhaps it's an "on the fly" temperature correction?
When you take a reading with the VST LAB refractometer, the instrument actually takes 30 separate measurements in a brief period of time. If the prism temperature varies more than 0.1C during this process, a "temperature stability error" is returned. You have to wait until things thermally stabilize.
This is distinct from "temperature compensation." That means you can make your measurements at any temperature between 15C and 30C. The result will be accurate as long as you (a) zero the instrument and (b) make your measurement at the same temperature.
Symbols: = New Posts since your last visit = No New Posts since last visit = Newest post
Forum Rules: No profanity, illegal acts or personal attacks will be tolerated in these discussion boards. No commercial posting of any nature will be tolerated; only private sales by private individuals, in the "Buy and Sell" forum. No cross posting allowed - do not post your topic to more than one forum, nor repost a topic to the same forum. Who Can Read The Forum? Anyone can read posts in these discussion boards. Who Can Post New Topics? Any registered CoffeeGeek member can post new topics. Who Can Post Replies? Any registered CoffeeGeek member can post replies. Can Photos be posted? Anyone can post photos in their new topics or replies. Who can change or delete posts? Any CoffeeGeek member can edit their own posts. Only moderators can delete posts. Probationary Period: If you are a new signup for CoffeeGeek, you cannot promote, endorse, criticise or otherwise post an unsolicited endorsement for any company, product or service in your first five postings.